Many customers have bought our 500 mA U-arm X-ray machine. After the sale, we will inform customers how to arrange the 500 mA U-arm X-ray machine room? The smallest area of the machine room for a 500 mA U-arm X-ray machine.
Briefly introduce to you the room conditions of the 500 mA U-arm X-ray machine
1. The installation foundation should be a cement foundation, and the ground should be solid, level, and capable of bearing 500 kg/m2;
2. The load-bearing walls on both sides should be able to withstand a load of 800 kg;
3. In order to effectively use the space behind the stereoscopic machine, the cable trench should be opened and the cable trench cover should be flush with the indoor floor. It is recommended that the trench section width * depth = 200 mm×150 mm, and moisture-proof, waterproof and rodent-proof measures should be taken in the trench;
4. Each room should be equipped with air-conditioning and dehumidification equipment in advance, and equipped with appropriate lighting and power sockets;
5. The requirements of protection must meet the national radiation health protection standards, and it is easy to use and not affected.
If you are interested in our product 500 mA U-arm X-ray machine, please contact us!