The customer consulted us about the X-ray machine for taking the lumbar spine. Through communication, we learned that the customer is currently using a 50mA X-ray machine, which is used to wash the film for imaging. The customer sent photos of the leg joint film and lumbar spine film. The customer asked how to take the lumbar spine. The shot is as clear as the leg joints. The client wants to perform lumbar spine radiographs for the patient and consult the improvement plan.
We explained to the customer that the 50mA X-ray machine can only take pictures of human limbs. If you want to clearly check the lumbar spine, you need to increase the shooting dose. It is recommended to choose a sickle arm X-ray machine with a value of 300mA or more. We provide customers with two improved solutions. One is to purchase a 300mA sickle arm X-ray machine with a digital imaging system. The second is to purchase a 500mA sickle arm X-ray machine, which can be used with existing film washing equipment. The digital imaging system has clearer imaging and can compensate to a certain extent for the lack of unclear imaging due to the low dose. The dose of the 500mA sickle arm X-ray machine is sufficient to clearly capture the human lumbar spine, including lateral imaging. The client approves of our plan, but the previous budget did not have that much, so we need to think about it again.