The Indonesian customer inquired about the double-column X-ray machine, and sent the information to the customer by WA
Double-pillar X-ray machine mainly includes ball tube column, chest film column, four-way floating camera bed, ground rail, electric
The gas control box is mainly used for supporting the medical diagnostic X-ray machine frame and the importance of mechanical movement
Components, convenient for the hospital to carry out gastrointestinal photography, gastrointestinal examination, chest and whole body bone photography
After the photography flat bed is assembled, the debugging is relatively simple. First, turn on the power supply of the two columns, and press
Lower the foot switch, release the flatbed locking electromagnet, and move the photographic flatbed bed plate (front, rear
Left, right) four directions, can move flexibly; Loosen the foot brake, and the bed plate cannot be moved. if there be
In case of poor locking, the gap of the locking electromagnet of the bed plate can be adjusted until it can be used normally.
When the camera flat bed film car is moving left and right, pull up the locking switch under the handle of the film car
It can easily move the tablet car left and right.
If you have other requirements for the double-column X-ray machine, please call us. Tel（whatsapp）:+86 17616362242