The sickle arm X-ray machine refers to the X-ray machine that can adjust the shooting angle and take pictures of the standing and lying positions of the human body. The basic components of the sickle arm X-ray machine include tube assembly, beam light device, high voltage generator, console, etc.
The sickle arm X-ray machine emits rays that penetrate the human body, receives the rays through the flat panel detector, and uses the automatic exposure detection function of the flat panel detector to convert the X-rays into images and display them on the computer for doctors to diagnose applications.
The maximum power of the sickle-arm X-ray machine is generally 30KW or 50KW. According to the different shooting parts of the human body, the power of the sickle-arm X-ray machine can be adjusted by adjusting the tube current and tube voltage of the X-ray machine. If you are shooting the limbs, chest cavity and other parts of the human body that are relatively easy to penetrate, you only need about 5KW of power; if you are shooting the thick and difficult to penetrate parts such as the lateral lumbar spine and pelvis of the human body, you need to increase the X-ray machine The voltage and current increase the power so that a clearer image can be taken.
If you are interested in sickle arm X-ray machine, you are welcome to contact us, phone (whatsapp): +8617616362243!